How to repair ICE122000787 Alma Lasers Soprano ICE Handle Piece?
Diode Laser Tech company have more than 10 years repairing service for alma lasers. We have also offered replacement and maintenance service for other brand like Syneron, Polarma, Cynosure and Nlight, Dilas etc.
But I think you may be interested in how we repair the ICE handle piece? So please read the following text carefully:
Firstly, When we receive an old ICE122000787 Alma Lasers Soprano ICE Handle, we will check the outlook carefully and take some detailed pictures as record for customer. We will take pictures from the shell, push bottom, water tube hose and the handle connector. After checking outside we will mark on the handle with customer name and sales name, date of receiving it.
Secondly, we will confirm with customer about the receiving of ICE122000787 Alma Lasers Soprano ICE Handle. Also will share with them the pictures we took from the ourside. If customer agree that we can open the diode handle to do the following replacement, then we will open it. Usually we open the shell, then take off the diode laser stack, disassemble the diode laser stack into pieces,and then check each bar under the microscope.
Thirdly, We will mount new diode laser bar onto heat sink and assemble diode laser stack to old structure in our cleaning lab. Both of our bar chips and micro channel cooler are imported from Germany manufacturer.We strictly manage mounting, assembling and testing step during laser stack producing. Every stack will be fully tested after more than 8 hours working in the lab.
Forthly, our experienced technician will reassemble the handle piece. Which include:
a. Install the new laser stack into the handle piece.
b. Clean lens and check the TEC and other parts inside the handle piece
c. Test the power output of whole handle piece.
d. Clean the cover shell of handle piece and package it
e. We send the finished good diode laser stack or hand piece to your address
Any more question please contact Danny whatsapp 0086-18518945778