Alma Lasers Soprano XL / ICE/ XLi/ Accord Alma Laser Tip Refurbishment details:
Alma Lasers Soprano XL/ICE/XLi/Accord Alma Laser Tip Refurbishment specifications:
|Application||Alma Soprano Handle Piece Repair And Replacement|
|Number of submounts||12 Bars|
|Power per submounts||100W|
|Central wavelength at 25℃||808 ± 5nm|
|Typical Operation current||60A|
|Typical Threshold current||15A|
|Maximum Operating Voltage||24V|
|Fast Axis Divergence (FWHM)||NO|
|Slow Axis Divergence (FWHM)||NO|
|Expected lifetime||>10000 Hours|
|Typical pitch per bar||1.8mm|
|Flow rate||0.3L/min ±10% / Bar|
|Water Temperature||15-25 ℃|
|Water Quality||Deionized 2-10 µs/cm, mixed bed ion exchanger,particle filter<25µm|
|cooling system||Do not use any material that combination with copper would form galvanic elements|
Alma Lasers Soprano XL / ICE/ XLi/ Accord Alma Laser Tip Refurbishment Stack Dimension
Alma Laser Tip Refurbishment handle piece and laser stack repair process:
First, the customers need to send us the old diode as below pictures shows, if can’t send old diode laser, need send to us structure design of the diode and dimension details stack. We will repair the old diode or design a new stack to replace the old one.
Second, we disassemble the handle piece, and test the old laser stack. For the micro channel laser stack, Such as Alam laser, Lightsheer, Syneron , we will test each laser plate one by one to confirm whether there are any one bars still can work. If there have some good bars, the repairing charge will lower. For macro laser stack like most product made in China, we will test whole laser stack together.
Third, we choose best quality laser stack to replace the old broken one. Our bars are imported from Germany. We mounted bars and assemble them to laser stack in our lab.Every stack will be fully tested and had 8 hours buring test.
Finally, our experienced technician will reassemble the handle piece. Which include:
a: Install the new laser stack into the handle piece
b: Clean lens and check the TEC and other parts inside the handle piece
c: Test the power output of whole handle piece
d: Clean the cover shell of handle piece and package it.
We can repair variety brand diode laser stack and lase handle.
Any more interest please kindly contact Danny whatsapp 0086-18518945778