3D NM Alma laser soprano ice platinum laser tip solution
If your 3D NM Alma laser soprano ice platinum laser tip need to be repaired, you can have the following 2 solutions:
FIRST solution is to replace old handle with new handle, oriental-laser can offer you a new handle without connector & I-bottom but with same power to match well.
This diode handle HP-Alma ICE is specially designed for 3D NM Alma laser soprano ice platinum laser tip Replacement. Engineers who want to repair 3D NM Alma laser soprano ice platinum laser tip or want to buy a new handle to replace old handle, you can buy a new handle to add Ibottom inside the handle connector to continue longer lifetime of the old machine.Also you should soldering old handle connector if the new connector not suitable or matched your old handle connector on the machine side,please find the details of picture following:
- Black Diode laser —
- Red Diode laser +
- Green+ White for ground-wire connect on contact
You should add this ground-wire in your side maybe
3D NM Alma laser soprano ice platinum laser tip Specifictaions
|Model No||3D NM Alma laser soprano ice platinum laser tip|
|Application||Alma Soprano ICE Laser Hair Removal Machine|
|Number of submounts||12 bars|
|Power per submounts||80W|
|Central wavelength at 25℃||808 ± 5nm|
|Fast Axis Divergence (FWHM)||NO|
|Slow Axis Divergence (FWHM)||NO|
|Expected lifetime||>10000 hours|
|Typical pitch per bar||1.8mm|
|Flow rate||0.3L/min ±10% / bar|
|Water Temperature||15-25 ℃|
|Water Quality||Deionized 2-10 µs/cm, mixed bed ion exchanger,particle filter<25µm|
|cooling system||Do not use any material that combination with copper would form galvanic elements|
3D NM Alma laser soprano ice platinum laser tip Structure
3D NM Alma laser soprano ice platinum laser tip Application
Also you can use it for old original laser alma soprano devices :
Second solution is to change the inner diode laser stack as new from the 3D NM Alma laser soprano ice platinum laser tip .
You may know that there are different inner diode block structure inside this kind of 3D NM Alma laser soprano ice platinum laser tip .
And the most common inner stack are the following A and B kinds, but we will not use original macro channel 12 bars to replace it, since macro channel lifetime is not good. But when we updated into 12bars micro channel , they can match well and working lifetime also better than before.
You may have A kind of inner stack, then we will use A block to replace like the following, they have same power with B block but with different water way only in order to match the water way and power for working.
While also you may have B kind of block inside your handle like this. The handle stack was fulled covered by the metal structure. Then we will use B kind of block to replace your old diode block.
If your laser stack like B block then we will use the same B block following to replace it. Anyway we have more than 10 years replacement of the ALma lasers. We know there are more than 5 kind of diode block inside the handle.But their power for matching and working are the same dear.
Experience For various brands diode laser handle repairing use
Any more interest for diode laser repairing service, please kindly add danny whatsapp 0086-18518945778.